NAVELEX EE 181-AA-OMI-050/E121 DT546-M9442
Table 5-1. Troubleshooting Procedures - Continued
c. Loose interconnect-
Ensure that all conduit and wire con-
nections are tight.
d. Gain or sensitivity
controls set too
BEFORE touching any gain or
sensitivity controls, note their
settings so that they may be
returned to their original posi-
tions after testing.
Check settings of Detector gain and
Processor sensitivity controls to
ensure that they are not set to