Table 2-1. Troubleshooting - Continued
TEST OR INSPECTION
c. Raise the theodolite sufficiently to disconnect the electrical connector (fig. 2-52)
from terminal (1A1J3) between thetbeodolite and GRU housing.
d. Complete removal of the theodolite by raising clear of the GRU housing.
Store the thedolite on a soft clean surface so as not to damage the theodolite mounting pads.
e. Install the upper band clamp adjusting tool (fig. 2-53), using two No. 4-40
by 3/8 inch attachment screws.