NAVELEX EE 181-AA-OMI-050/E121 DT546 M9442
Table 5-1. Troubleshooting Procedures - Continued
that secure PC board to chassis. Lift
PC board out of chassis.
(2) To install new PC board, orient board
so the terminal strips are adjacent to
conduit entrance holes. Secure the
PC board to chassis with screws.
Connect wires to TB1 and TB2. Adjust
sensitivity control per instructions in
If PC board is good, check pick-up
d. Bad Detector pick-
assembly in Detector. Disconnect red
and black leads from PC board. Set
multimeter to ohms (highest scale), and
connect meter leads to red and black
Meter should indicate over
If this indication is not present, replace
disconnect red and black leads from
PC board. Remove screws that secure
assembly to mounting surface, and
To install new pick-up assembly, orient
assembly so that red and black leads
are toward PC board. Tilt assembly so
that edge nearest PC board can be
slipped under PC board.
assembly into chassis, flat against
mounting surface, and secure it with
screws. Connect red and black leads
to PC board.
BEFORE touching any gain or sensitivity
controls, note their settings so that they may
be returned to original positions after testing.
e. Gain or sensitivity controls
set too high.